Electronic scale error in excess of the prescribed solution
Electronic scales are used to weighing the goods of the weight of an electronic scale.
Often use electronic scale users must clear, the scale will appear some error is normal, but when the electronic scale error more than the provisions of the state, that is about to snack on.
Today, electronic scale small make up to you to sort out this electronic scale error more than the provisions of the state solution.
Hope to be able to help to you!
The main cause of electronic scale error than the provisions of the state are: electronic scale between the balance of body and foundation have sundry, each sensor Angle with horns is poor, the spacing device of electronic weigher body scale basis, and some because of equipment failure will also cause out-of-tolerance condition, down to see the corresponding solution.
Electronic scale error in excess of the prescribed solution: a, electronic scales between the balance of body and foundation have sundry, this kind of situation.
Use because some customers usually neglect cleaning and maintenance, lead to debris in the balance of body and foundation cracks and under the scale, because we know that the electronic scale pressure induction principle is to use the sensor, and sundry and scale body contact will cause the change of the stress, resulting in the change of the weighing data, leading to electronically produced by the increasing the scale of error out-of-tolerance condition.
So electronic scale of daily maintenance and cleaning of electronic scales of precision and accuracy is also very important.
Second, each sensor Angle with horns.
An electronic scale is the support of more than four of the weighing sensor, and the force situation of every Angle is not consistent, so in the installation and debugging of electronic scales need to debug the Angle difference in the junction box or instrument, if you don't adjusted the Angle difference, there will be such as trucks to touch a different position before and after will appear different weight, we also call it the Angle difference, same Angle difference is allowed within a certain range, such as the value of a degree.
A poor corner of debugging good electronic scale, using a gears, Angle difference is also likely to change, so it's best to six months or a year for the calibration.
Three, the spacing device of electronic scale scale body base, electronic scale co. , LTD mostly institutions, is to limit the electronic scale on a fixed position, in order to avoid shifting when car weighing electronic scale, but the scale of the limit screw without touching base again, otherwise will affect the weighing values, again so adjust spacing generally is about 3 mm, in the winter and summer to do inspection.
And some because of equipment failure will also cause out-of-tolerance condition, there would not be in do blieve!
Through the above understanding, everybody on this electronic scale error over the regulations of the state, the solution of the article is clear.
If you still have what not understand of place, welcome to inquire!
In this paper, by the Shanghai heng electronic scale small make up finishing just instrument co. , LTD. , thank you for your support for our company for a long time!
Partners also like to read this article: use electronic scales the note!